Insertion Type Ultrasonic Liquid Flow Meter – EUF1510I
- Meet Standard of JJG 1030-2007 for ultrasonic flow meter
- Approved by ISO9001
Hurricane EUF1510I Flow meter with accuracy 1%
Hurricane EUF1510I Flowmeter –Insertion type Ultrasonic Liquid Flow Meter consists of a flow sensor and a flow transmitter.
The EUF1510I is a state-of-the-art universal transit time Flowmeter designed using advanced digital technique, intelligent adaptive acoustic technology and low-voltage broadband pulse transmission. While principally designed for clean liquid applications, the instrument is tolerant of liquids with the small amounts of air bubbles or suspended solids found in most industrial environments.
With features of small-size, insertion type, easy installation, module type, rechargeable battery powered, and universal power supply, Hurricane EUF1510I work for measuring Liquid Flow in tubes anytime anywhere.
Standard Configuration of EUF1510I
1. Flow Transmitter 2. Flow Sensor
3. Brass ball valve 4. Carbon steel base
♠ High quality signal receipt. It adopts the sound focus patent technology to make the quality of signal reception more obviously strengthen in continuously measurement.
♠ Outstanding high reliability. By using the advanced digital technique, intelligent adaptive acoustic technology, the reliablity of Hurricane EUF1510I Flowmeter’s measure is more outstanding.
♠ With distinctive features such as high precision, low consumption power, anti-jamming, outstanding applicability, high capability and low cost, the EUF1510I Flowmeter features other advantages.
♠ Clear, user-friendly menu selections make EUF1510I Flowmeter simple and convenient to use.
♠ Daily, monthly and yearly totalized flow: Totalized flow for the last 64 days and months as well as for the last 5years are may be viewed.
♠ Standard flange installation. Very easy installation make the use easier and the measure Liquid Flow in tubes more flexiblely. The measure can be done after the electric connects well.
♠ Easy installation. Online with power installation make the measure Liquid Flow in tubes more flexiblely.
♠ Hurricane Ultrasonic Liquid Flow Meter – EUF1510I gathers the RS485 communication interface, 4-20mA current signal, pluse output and Modbus protocol.
♠ Hurricane Ultrasonic Liquid Flow Meter – EUF1510I can expand the solar power, anti-thunder power and anti- explosion or other function modules, so the EUF1510T flowmeter can be used in various environment.
- Carbon steel
- Stainless steel
- Cast iron
- Cement pipe
- Aluminum and others
- Running water
- Sewage treatment
- Chemical material
- Crude oil
- Sea water desalinization
- Lubricating oil
- Industry Circulating Water
- Chemical reagent adding and others
- Water, sewage (with low particle content) and seawater
- Water supply and drainage water
- Power plants (nuclear power plant, thermal and hydropower plants), heat energy, boiler feed water and
- Energy management system
- Metallurgy and mining applications (cooling water and acid recovery, for example)
- Petroleum and chemicals
- Food, beverage and pharmaceutical
- Marine operation and maintenance
- Energy economy supervision and water conservation management
- Pulp and paper
- Pipeline leak detection
- Regular inspection, tracking and collection
- Energy measuring and balance
- Network monitoring systems and energy/flow computer management
Connection mode & Dimensions
⇒ Connection diagram
⇒ Dimensions of transmitter unit: mm
The size of sensor
It is very easy to install the ultrasonic flow meter. Choose one suitable checking point, set the pipe parameter on the flow meter, and fix the transducer on the pipe, and then the flowmeter works. Please take reference on the rules as below while installing.
The installation of ultrasonic flow meter is the most simple and quick in all flow meter. Only choose one suitable checking point, input the pipe parameter to the flow meter and fix the transducer on the pipe then can make the measurement. Please take reference on the rules as below while installing.
Checking-point select principles
1.Get avoid from the position point 6 and 12 at o ‘clock direction to avoid signal attenuation due to the sediment at the bottom of the pipe or the bubble and cavitations.
2.Adequate consider the deposit condition within the pipe, try to choose the part without deposit for measurement. When completely meet the conditions, it need to take deposit into consideration as lining for better measurement accuracy.